Power through a small diameter
An angiographic or diagnostic catheter is intended for use in angiographic procedures. It delivers radiopaque media and therapeutic agents to selected sites in the vascular system. It is also used to lead a guidewire or a catheter to the target site.
-The thin wall structure permits flow rates up to 23 ml/sec (saline) and a maximum pressure limit of 1,200 psi (8.274 kPa)1
-Braiding with variable pitch offers partial proximal rigidity and increased flexibility at the distal end2
-The Tiger and BLK curves enable angiography of right and left coronary arteries using either a radial or brachial approach. This single catheter concept decreases procedure and fluoroscopy times, providing both procedural and cost-efficiency3
4 Fr (1.40 mm) – 1200 psi
Outer diameter of the shaft
4 Fr / 1.40 mm
1 Terumo Instructions for Use, Outlook catheter.
2 Data on file. Ref TAC-001.
3 Seong-Man Kim et al, 2005. Novel Diagnostic Catheter specifically designed for both coronary arteries via the right transradial approach. The International Journal of Cardiovascular Imaging. 2005, 10.007/s10554-005-9029-8.